Send the link below via email or IMCopy
Present to your audienceStart remote presentation
- Invited audience members will follow you as you navigate and present
- People invited to a presentation do not need a Prezi account
- This link expires 10 minutes after you close the presentation
- A maximum of 30 users can follow your presentation
- Learn more about this feature in our knowledge base article
Do you really want to delete this prezi?
Neither you, nor the coeditors you shared it with will be able to recover it again.
Make your likes visible on Facebook?
You can change this under Settings & Account at any time.
Transcript of MFD-MICROMAX PLUS
MONTCLAIR FIRE DEPARTMENT
MICROMAX PLUS/Series One
-FOUNDED IN 1906 BY MARK C. HONEYWELL
-COMPANY MERGED WITH "Minneapolis-Honeywell Regulator Company", FIRST COMPANIES TO SUPPLY "DAMPER-FLAPPER THERMOSTATS"
-IN WWII, HONEYWELL JOINS THE DEFENSE DEPARTMENT WHICH STARTED THE AREOSPACE DEPT.
-FIRST TO DEVELOPE DOMESTIC CO MONITORING
Programbable gas detector
-reads up to 5 gases with combo-tox sensor
-Liquid Crystal Display
-high contrast 4 lines by 16 aplhanumeric display
-automatic fiber-optic backlight
-battery can be used as NiCAD rehchargable pack or with 2 "AA" = 9 hours of operation (4 "AAA" for st. 2)*
-low battery alarm is double beep every 15 seconds
- *for micromax series one, 1 beep every 15 seconds until 1.9v is reached, then continuous alarm until unit shut down.
-Confidence Beep every 20 seconds, if programed, to insure instrument is working.
-battery gage, if programed, with up to 6 Bars
-Audible and Visual Alarms
-low flow (steady audible alarm with visual alarm)
-display indicates low flow
-RFI (radio frequency interference) minimizes false alarms
-Storage = 200 hours of data
- -4 to 122 degress F operation tempurature.
- 90 db at 1 ft
- LESS THEN 19.5%
- BECOME HYPOXIC
ALOC WITH MEMORY LOSS
DEATH MAY BE RESULT
Originally incorporated as EIC-Sieger in 1959 and later re-branded as J&S Sieger Ltd in 1961.
- Joshua Sieger invented the first low-power catalytic bead in 1958, designed for the detection of flammable gases onboard boats.
-Sieger sells company to Zellweger Analytics in 1979
-Zellweger is bought out and rebranded in 2005 by Honeywell; Honeywell Analytics
- Turn unit on by pressing ON/OFF button
-wait to hear confirmation beep.
-instrument will display firmware version #
-Instrument will go through "Testing" and "Zeroing" cycles, displays low-alarm and High-alarm levels
-indicates "ready" and goes into operation mode.
Different standard modes are available by pressing "MODE" button
-PEAK READING;(hightest toxic and combustible gas levels and the lowest o2 levels)
- STEL (short term exposure limit) values for toxic gases.
-TWA (time weighted average) for O2, LEL, Toxic gases.
-10' tube used for confined space or remote areas
-Insure instrument is in "P" pump mode
-connect hose to monitor
-place in space to be monitored and wait 2 minutes
-> may use up to 100' hose; add 1 sec. per ft added
-> hose should mainly be used for pre-sampling; may drain battery.
-If gas alarm is present evacuate if you do not have proper PPE.
-> low gas alarm = 3 beeps/red flashing light/flashing gas range on monitor.
-> high gas alarm = 3 beeps/flashing light/flashing gas range
->Low flow alarm= continuous audible alarm/ flashing LED with words "LOW FLOW"
-> Battery Shutdown = 5 audible beeps.
OXYGEN (O2) = 19.5% to 23.5%
COMBUSTIBLE (%LEL) Methane = 10% to 20%
CARBON MONOXIDE (CO) = 25PPM to 35PPM
HYDROGEN SULFATE (H2S) = 10PPM to 20PPM
-BLOOD BECOMES ALKALOTIC
-CELL DAMAGE MAY OCCURE
-SIGNS AND SYMPTOMS
-COLORLES, ODERLESS GASS PRODUCED BY BURNING MATERIAL CONTAING CARBON
-LEADING CAUSE OF ACCIDENTAL DEATH IN AMERICA
-SIGNS AND SYMPTOMS
-HEADACHE, NAUSEA, FATIUGE
-SOB, C.P., N/V.
-DEATH BY OXYGEN DEFICIENCY.
- 1,500 ppm is fatal
NOT A TOXIC SUBSTANCE STANDING ALONE; THOUGH HIGHLY FLAMABLE.
-MAY CAUSE ASPHYXIATION IF IN A CONFINED SPACE OR AREA WITH POOR VENTILATION.
-CAN BE INTRODUCED TO BODY BY: INGESTION, INHALATION, AND TOUCH.
-MAIN COMPONET OF NATURAL GAS.
-SIGNS AND SYMPTOMS
-CNS DEPRESSENT (HYPOXIA)
COLORLESS GAS WITH "ROTTEN EGG" LIKE SMELL.
- FOUND FROM HOT SPRINGS, UNDERWATER THERMAL VENTS.
-LESS THAN 50PPM - EYE IRRATION WITH MILD HEAD ACHE
HIGH LEVELS OF H2S
- > 100PPM, MAY NOT DETECT ROTTEN EGG SMELL.
-> S/S MAY NOT SHOW UP UNTIL 72 HOURS
- >1,000PPM WILL CAUSE DEATH WITHIN 30 MINUTES.
**must be done in clean air environment.
**insure that calibration gas is within expiration date.
- Use mode button to scroll to "user set up"
- depress "ON/OFF" button to enter
-instrument will count down to 0
-apply calibration gas.
-instrument will display "CALIBRATING"
-all sensors must be displaying post "CAL COMPLETE"
-instrument goes into purging countdown to 0
-memory is cleared.
-use hex wrench to remove calibration screw
-turn on monitor.
-use wrench to tap switch inside port once.
-wait 30 seconds then apply gas; apply gas appropriate to instruments sensor complement.
-before removing gas, tap calibration screw again, then gas levels will appear on display.
TWO TYPES OF CALIBRATION TIME FRAMES
- "bump test"
-exposes sensors to known concentration test gas every day
-calibration interval can be every 90 days.
- If "Bump Test" is not performed, check monitor every few days to insure nothing in you atmosphere is damaging sensors (high concentrations of combustible gas, terta-ethel-lead, sulfides or silicone lubricants).
-if this is done calibration interval is every 30 days.
- recalibrate if measurements are off by 15%.