Send the link below via email or IMCopy
Present to your audienceStart remote presentation
- Invited audience members will follow you as you navigate and present
- People invited to a presentation do not need a Prezi account
- This link expires 10 minutes after you close the presentation
- A maximum of 30 users can follow your presentation
- Learn more about this feature in our knowledge base article
Do you really want to delete this prezi?
Neither you, nor the coeditors you shared it with will be able to recover it again.
Make your likes visible on Facebook?
Connect your Facebook account to Prezi and let your likes appear on your timeline.
You can change this under Settings & Account at any time.
Transcript of wireline schlumberger
My work experience :
Dosty M. Jafar
- tools and their calibration and tools used to check or fix them like trim machine manometer , test box , evacuating pump ..
- rig up
- pipe recovery
- IDW and CMTD
How did we spend the week of field exposure ?
What’s TLD ?
Three detector lithology density tool which contain of three part
HRGD ( source and three detector) – HRMS (sonde) –
HRCC (catridge )
. Gamma rays emitted from radioactive source
. Gamma rays collide with electrons in formation losing energy
. Detectors measure intensity of backscattered gamma rays.
-- high energy GR relate to Density
-- Low energy GR relate to Lithology
Principle of operation :
Back scatter detector (BS)
Long spacing Detector (LS)
short spacing Detector (SS)
1. bulk density :
The bulk density in g/cc is the weighted sum of the matrix and fluid densities.
The individual characteristics of a rock in terms of its mineral composition and structure
3. borehole diameter:
The diameter of the borehole is measured from the face of the skid pad to the end of the caliper arm that holds the skid against the formation
Comparison between Master Calibration and Wellsite Stabilization measurements. Difference must be within tolerance
Repeatability and accuracy:
• The calibration is performed every 1 month or after every 20 jobs.
• The calibration consists of measurements of the background signal from the stabilization source.
• Measurements of the count rates are made in the GCB-L aluminum block and the GCB-M magnesium block.
• Measurements are performed twice, once with and once without a 3-mm stainless steel sleeve.
•Calibration Quality Control Listing gives Mean & Maximum Deviations and Tolerances for the three detectors
•Values are flagged if out of tolerance.
TLD Master Calibration
1. fit 1 and fit 2:
3. op check
5. equipment used in operations:
Accuracy, is measuring once and having a reliable reference to check accordingly.
Repeatability is measuing repeating the measurement steps for many times till the result averages out.